A Nicol prism is a type of polarizer, an optical device used to produce a polarized beam of light from an unpolarized beam. It is made in such a way that it eliminate one of the ray by Total Internal Reflection i.e, O-ray is eliminated and only E-ray is transmitted through the prism.See polarized light. It was the first type of polarizing prism to be invented, in 1828 by William Nicol (1770–1851) of Edinburgh. It consists of a rhombohedral crystal of Iceland spar (a variety of calcite) that has been cut at an angle of 68° with respect to the crystal axis, cut again diagonally, and then rejoined as shown using, as a glue, a layer of transparent Canada balsam.
Unpolarized light enters through the left face of the crystal, as shown in the diagram, and is split into two orthogonally polarized, differently directed, rays by the birefringence property of the calcite. One of these rays (the ordinary or o-ray) experiences a refractive index of no = 1.658 in the calcite and it undergoes total internal reflection at the calcite-glue interface because its angle of incidence at the glue layer (refractive index n = 1.55) exceeds the critical angle for the interface. It passes out the top side of the upper half of the prism with some refraction as shown. The other ray (the extraordinary ray or e-ray) experiences a lower refractive index (ne = 1.486) in the calcite, and is not totally reflected at the interface because it strikes the interface at a sub-critical angle. The e-ray merely undergoes a slight refraction, or bending, as it passes through the interface into the lower half of the prism. It finally leaves the prism as a ray of plane polarized light, undergoing another refraction as it exits the far right side of the prism. The two exiting rays have polarizations orthogonal (at right angles) to each other, but the lower, or e-ray, is the more commonly used for further experimentation because it is again traveling in the original horizontal direction, assuming that the calcite prism angles have been properly cut. The direction of the upper ray, or o-ray, is quite different from its original direction because it alone suffers total internal reflection at the glue interface as well as a final refraction on exit from the upper side of the prism.
Nicol prisms were once widely used in microscopy and polarimetry, and the term "using crossed Nicols" (abbreviated as XN) is still used to refer to the observing of a sample placed between orthogonally oriented polarizers. In most instruments, however, Nicol prisms have been replaced by other types of polarizers such as Polaroid sheets and Glan–Thompson prisms.
Famous quotes containing the word prism:
“Neither can I do anything to please critics belonging to the good old school of projected biography, who examine an authors work, which they do not understand, through the prism of his life, which they do not know.”
—Vladimir Nabokov (18991977)