The anaesthetic machine (UK English) or anesthesia machine (US English) is used by anaesthesiologists, nurse anaesthetists, and anaesthesiologist assistants to support the administration of anaesthesia. The most common type of anaesthetic machine in use in the developed world is the continuous-flow anaesthetic machine, which is designed to provide an accurate and continuous supply of medical gases (such as oxygen and nitrous oxide), mixed with an accurate concentration of anaesthetic vapour (such as isoflurane), and deliver this to the patient at a safe pressure and flow. Modern machines incorporate a ventilator, suction unit, and patient monitoring devices.
The original concept of Boyle's machine was invented by the British anaesthetist Henry Boyle (1875–1941) in 1917. Prior to this time, anaesthetists often carried all their equipment with them, but the development of heavy, bulky cylinder storage and increasingly elaborate airway equipment meant that this was no longer practical for most circumstances. The anaesthetic machine is usually mounted on anti-static wheels for convenient transportation.
Simpler anaesthetic apparatus may be used in special circumstances, such as the TriService Apparatus, a simplified anaesthesia delivery system invented for the British armed forces, which is light and portable and may be used effectively even when no medical gases are available. This device has unidirectional valves which suck in ambient air which can be enriched with oxygen from a cylinder, with the help of a set of bellows. A large number of draw-over type of anaesthesia devices are still in use in India for administering an air-ether mixture to the patient, which can be enriched with oxygen. But the advent of the cautery has sounded the death knell to this device, due to the explosion hazard.
Many of the early innovations in U.S. anaesthetic equipment, including the closed circuit carbon-dioxide absorber (aka: the Guedel-Foregger Midget) and diffusion of such equipment to anaesthetists within the United States can be attributed to Richard von Foregger and The Foregger Company.
In dentistry a simplified version of the anaesthetic machine, without a ventilator or anaesthetic vaporiser, is referred to as a relative analgesia machine. By using this machine, the dentist can administer a mild inhalation sedation with nitrous oxide and oxygen, in order to keep his patient in a conscious state while depressing the feeling of pain.
Other articles related to "anaesthetic machine, machine, machines":
... The Anesthesia machine contains mechanical respiratory support (ventilator) and O2 support as well as being a means for administering anesthetic gases which may be used for sedation as well as total ...
... The anaesthetic machine (UK English) or anesthesia machine (US English) or Boyle's machine is used by anaesthesiologists, nurse anaesthetists, and ... The most common type of anaesthetic machine in use in the developed world is the continuous-flow anaesthetic machine, which is designed to provide an accurate and continuous supply of medical gases (such as ... Modern machines incorporate a ventilator, suction unit, and patient monitoring devices ...
... Based on experience gained from analysis of mishaps, the modern anaesthetic machine incorporates several safety devices, including an oxygen failure alarm (aka 'Oxygen Failure Warning Device' or ... In older machines this was a pneumatic device called a Ritchie whistle which sounds when oxygen pressure is 38 psi descending ... Newer machines have an electronic sensor ...
Famous quotes containing the word machine:
“I find it hard to believe that the machine would go into the creative artists hand even were that magic hand in true place. It has been too far exploited by industrialism and science at expense to art and true religion.”
—Frank Lloyd Wright (18691959)